Even if it is a very good surgical bed, there may be some failures in the use, and because the treatment of patients is very time -effective, users often need to solve some small failure problems by themselves to avoid the failure of the bed due to the bed failure. The problem brings inconvenience to surgery. The so -called user needs to solve some small failures by themselves. It means that when the surgical bed does not have small problems such as forward motion, users can deal with themselves. From the perspective of this aspect, everyone first needs to observe whether it is just a forward action, and other actions are available. If so, it means Failure or corresponding solenoid valve.
After the preliminary judgment of the above, users need to determine whether the solenoid valve of the surgical bed is good or bad. There are two methods that can be used in this diagnosis. See if there is any inhalation, if the diagnosis determines that the solenoid valve is normal, it is necessary to consider whether it is caused by the obstruction of the oil circuit. However, if the surgical bed is not just without the front, and other actions are not available, it is usually a compressed pump failure. For this kind of situation, you need to check whether there is voltage on the shrinkage pump. Whether the resistance is normal. If these are normal after checking, it means that it is caused by the failure of the capacitance.